- The CACTI campaign has ended and the radar is being packed.
Joseph C. Hardin
Radar is installed at AMF1 site for CACTI. It is operating and collecting data.
- This radar is at LANL and will not be deployed for CACTI.
The LASIC campaign has ended. The radar has been packed for return to the US.
- Site operations restarts the radar when it sporadically stops operating.
- Pedestal servo amplifier went through extensive failure analysis and the vendor is going to ship the repaired servo amplifiers. Based on the analysis, it appears that faulty capacitors were used in production. Servo amps will be replaced in all pedestals with the repair performed as we get the repaired ones back from the vendor.